Executable .exe-Files
- laser_ket - Laser
- ut75_ket1 - Temperature sensor

Information about Temperatur sensor with KET:
- simply execute .exe-File 
- check dB-Lab at Hardware -> KET -> KET External Sensor Monitor
- press CTRL + C to quit

Information about Laser with KET: 
- Follow the enclosed application note for setting up the temperature sensors

More Information: 

Supported Devices:
- One sensor head from LKG5000 laser series from Keyence.
- Two processing channels (peak, bottom) are used

LKG5000 Specifications:
- Standard control unit supports 2 heads with corresponding output (processing) channels.
- Expansion units are available to allow support up to 12 heads
  (not supported by this script).
  Depending on selected processing channels per head not all heads can be used.
- Only one control unit can be controlled by a computer.

Limitations:
- Currently supports only one laser head.
- Requires two output processing channels for peak and bottom measurements.
- Dynamic DC offset is an numerical mean of peak and bottom, not a averaged
  calculation of time signal, which might be more appropriate.

Future Development:
- Support for up to 6 laser heads with Head expansion units for separated
peak and bottom measurements.
- Support for up to 12 laser heads with additional expansion units
and peak-to-peak measurements.
- Support for other laser series.
 